The device uses the method of ultrasound examination of the paranasal sinuses, based on the property of ultrasound reflected at the boundary of two media with different acoustic resistance. Given the direction of the beam and the position of the reflecting point, it is possible to determine the location of the structures of interest to the researcher. Echosinuscopy considers echo signals reflected from the boundaries of "soft tissue/bone", "bone/air", "bone/sinus contents", "sinus contents/bone".
The reflection coefficient of ultrasound is very different for different interfaces. At the "bone/air" boundary, there is a complete reflection of ultrasound. At the "bone/sinus contents" boundary, a partial reflection of ultrasound occurs, and part of the ultrasonic energy passes through the boundary and spreads in a liquid, for example, in the liquid contents of the maxillary.
|Display||Color LCD display with a screen resolution of 400*240|
|Two modes operation||A-scan & B-scan (A-scan - displaying the signal envelope, determining the distance to the reflection boundary, B-scan - the color gradient of the signal change over time)|
|Transducer||Ultrasonic crystal 2.5±0.5 MHz|
|Depth of probing||not less than 80 mm|
|Internal storage||2 GB|
|USB connection to a PC for saving and printing data||Available|
|Precision||not worse than 1 mm|
|Resolution capability||0.3 mm|
|Automatic and manual mode (gain mode)||Available|
|Automatic markers of the distance to the reflection boundary||Available|
|Peak value negative sound pressure||not more than 1 MPa|
|Output beam intensity||no more than 20 MW/cm2|
|Spatial-peak temporal intensity||not more than 100 MW/cm2|
|The time of setting the operating mode (the device is ready for operation)||no more than 10 seconds|
|Continuous operation time of the device||at least 8 hours|
|The time of automatic shutdown of the device in the absence of a signal (or in the absence of pressing the buttons)||5±1 minutes|
|Power supply||AC 220 V ± 10%, 50 Hz
built-in rechargeable battery with a capacity of 1250 mAh
|Power consumption||no more than 10 W|
|Overall dimensions||250 x 185 x 60 mm|
|Weight with accessories||1,485 kg|
|There is no indication or indication between 0 and 1.5 cm. Negative symptomatic.||Fluid in the maxillary sinus. The display is between 3.5 and 6 cm. Positive symptomatic.||Cyst or polyp in the maxillary sinus. The display is between 2 and 3 cm.|
|Fluid in the frontal sinus. The display is between 1 and 3 cm. Positive symptomatic.||Fluid in the maxillary sinus of the child. The display is between 2 and 4 cm. Positive symptomatic.|
Do you have a question?
Call our specialist or send us a message.